[sample, raw] = Eyelink(‘NewestFloatSampleRaw’ [, eye])
Same as Eyelink(‘NewestFloatSample’) but additionally returns raw sample data.
Specify the eye for which raw values are desired as LEFT_EYE or RIGHT_EYE as
returned by EyelinkInitDefaults. Omission of this argument is deprecated.
Normal samples do not contain the corneal reflection data, but some
(non-saccade-based) calibration methods require this information.
The Eyelink 1000 can be configured to send this information as part of ‘raw’
Sol Simpson at SR-Research emphasizes that this is not officially supported or
- You may need to install Eyelink.dll from the latest software developer kit at
https://www.sr-support.com/forums/showthread.php?t=6 (windows) or
- Issue the commands:
Eyelink(‘command’,’inputword_is_window = ON’);
HMARKER (originally for Eyelink2’s infrared head tracking markers) and INPUT
(originally for the TTL lines) are jury-rigged to hold the extra data.
You can also set file_sample_data to collect raw samples in the .edf file.
Raw structure fields:
raw_pupil raw x, y sensor position of the pupil
raw_cr raw x, y sensor position of the corneal reflection
pupil_area raw pupil area
cr_area raw cornela reflection area
pupil_dimension width, height of raw pupil
cr_dimension width, height of raw cr
window_position x, y position of tracking window on sensor
pupil_cr calculated pupil-cr from the raw_pupil and raw_cr fields
cr_area2 raw area of 2nd corneal reflection candidate
raw_cr2 raw x, y sensor position of 2nd corneal reflection
CAUTION: It may or may not work on your setup with your tracker.